Fluke Corp., Everett, Wash., is voluntarily recalling Fluke 373, 374, 375 and 376 digital clamp meters manufactured between September 1, 2010, and October 31, 2012. According to the company, the printed circuit assembly in these units may not be properly fastened to the test lead input jack. This may result in inaccurate voltage readings, including a low or no voltage reading on a circuit energized with a hazardous voltage, presenting a shock, electrocution or thermal burn hazard.
|Model||Manufacture Dates||Starting Serial Number||Ending Serial Number|
|Fluke 376||Sept 2010-Feb 2011||14270001||15909999|
|Fluke 375||Sept 2010-Mar 2011||14270001||16079999|
|Fluke 374||Oct 2010- Feb 2011||14270001||16379999|
The voluntary recall involves the digital clamp meters with the model and serial numbers listed at right. Any serial number preceded or followed by the letter “R” is not affected by this recall notice.
If you own one of these clamp meters, the company asks that you stop using it immediately and send it back to Fluke for repair. To contact Fluke for information on how to have your units repaired, either visit www.fluke.com/37x-recall or call (877) 564-7180.